Level III Certified Infrared Thermographer®is a three-day course focused on best practices for infrared inspections and associated activities. This course contains information vital to the long term success of an infrared inspection program.
This course covers advanced topics related to developing, implementing, and managing a successful infrared inspection program. Topics include latest applications, hardware and software, current industry standards and specifications, OSHA and NFPA safety standards, marketing and promoting an infrared inspection program, thermography as legal documentation, heat transfer analysis software, current industry certification criteria, and how to develop and implement standards-compliant written practices and procedures.
Students are encouraged to bring their own imager for individualized training or to learn how to use it more effectively.
Course tuition includes all course presentations, Student Reference Manual, and Infraspection Institute Certified Infrared Thermographer®exam. Certification card and diploma issued with a passing grade of 80%.
The course is approved by the InterNational Association of Certified Home Inspectors for 21 hours of continuing education and meets the training requirements for their Infrared Certified professional designation and logo.
Prerequisite: Level II Thermographer Certification
Level III Thermographer Training Curriculum
1. Ensure a Successful Infrared Inspection Program
Define a successful program and identify program obstacles
Understand the requirements of OSHA, NFPA, and other regulatory agencies
Identify and specify safe practices for performing IR inspections
Eliminate recurring problems through the application of Root Cause Failure Analysis
2. Explore Latest Technology
Discover newest applications for infrared thermography
Discuss the features and benefits of new test equipment
Utilize state-of-the-art software for data management & report generation
Identify industry trends and business opportunities
3. Understand Industry Standards & Specifications
Review current standards and specifications
ASTM, ANSI, ASNT, Infraspection Institute, ISO, NETA, NFPA
Examine current NDT certification programs
Identify certification criteria
Develop a written practice for thermographer certification using ASNT standards
4. Marketing and Promoting an Infrared Inspection Program
Enlist principles of marketing to identify, position and price IR services
Analyze best practices for effective promotion
Discuss key elements for an effective website
Explore e-commerce opportunities to increase revenue
5. Thermography and the Law
Review the importance of thermography as legal evidence
Examine the role of the thermographer as an expert witness
Identify appropriate report content
Learn cardinal rules for delivering testimony
6. Applying Tools to Ensure Program Longevity
Learn five methodologies for calculating program payback or ROI
Understand how to integrate thermography into a Quality Management System
Discover how to effectively identify and solve problems
Identify software that can slash report writing time by more than 50%
7. Integrating PdM Technologies
Discuss role of thermography as a PdM technology
Identify complementary PdM technologies
Learn necessary steps to effectively implement a PdM program
Ensure program longevity through proper planning and management
FAQ - Fotric 226B and 223B AI temperature screening thermal imager camera scanner
Here is the list of possible issues you may encounter and the solutions/suggestions for your Fotric 226B and 223B AI group temperature screening thermal imaging camera and Fotric WLIR software operation.
1. Backlighting problem (cannot detect human face)
The recommended installation environment for the FOTRIC 226B and 223B is indoors with good lighting. Try to avoid a backlit environment, which can affect the accuracy of face detection or even result in detection failure.
When the device is facing a door or window, the facial detection failure in the figure below may occur.
It is recommended to change the orientation of the device or turn on the indoor lights to reduce the effect of backlighting.
2. Accurate focus
The FOTRIC 226B and 223B use manual focus adjustment, and it can only correctly measure temperature if its focus is accurate. If the focus is not accurate, the infrared image can be blurred, and the corresponding measured temperature can also result in a large error.
Blurred focus and inaccurate temperature
Accurate focus and accurate temperature
To use the device correctly, the visible and infrared images must be matched.
The picture below shows an unmatched photo. It can be seen that the visible light detection frame (right) is positioned to the face, while the infrared detection frame (left) does not match the face position. The temperature test result obtained is incorrect, as it has not read a human face temperature.
After matching, the face detection frame of the infrared photo and the visible light photo are in the same position, so the measured result is accurate
4. Threshold mode
The threshold mode directly displays the human body surface (face) temperature. The body surface temperature is greatly affected by the environment.
In threshold mode, the temperature compensation can be performed manually, and the compensation temperature value is ≤10 °C.
5. HawkAI Mode
The working principle of the HawkAI mode is that the FOTRIC 226B and 223B first collects the human body surface (face skin) temperature, and then automatically calculates the compensation temperature through the AI algorithm to convert the human face skin temperature into the real body temperature. HawkAI mode can adjust the compensation temperature value in real time according to the flow of people continuously passing the temperature measurement point, so as to adapt to the change of ambient temperature.
Please note: The AI mode is designed with caution. The goal is to screen out people with a high body temperature in the crowd, and then perform a second temperature measurement. Body temperature varies from person to person; the biggest difference can be as high as 1-2 °C. With careful consideration, please have the screened person to stabilize for 5-10 minutes, and then perform a second temperature measurement. If their temperature is still high, it is recommended to use a medical thermometer to repeat the measurement.
6. HawkAI mode data sampling
When using AI mode, the software will prompt: “You have to resample temperature for 10 sets”. We recommend sampling the temperature of 10 different people with variations, so that the temperature compensation value of the AI algorithm can fit most people.
7. No alarm above 40 ℃
When the measured temperature of the human face exceeds 40 °C, the software will not emit an alarm. The software’s upper limit of the alarm temperature is 40 °C. If the temperature of the human body exceeds 40 ° C, the person would feel unmistakably ill and should go to the hospital for treatment immediately. If the person keeps appearing in some places, he or she may use many methods to reduce his or her face temperature to prevent being detected by infrared thermal imaging devices.
8. The effect of distance on temperature measurement
First, we recommend that the location of the temperature measurement point is fixed, and the distance from the device to the measurement point is also fixed. Then the device is focused and matched under this premise. Therefore, when the person to be inspected reaches the temperature measurement point, the focal length is appropriate, and the measured temperature is accurate.
If the person being inspected continues to walk towards the IR imaging camera, he/ she will be closer to the infrared camera, the focal length of the infrared image changes, and the measured temperature may increase.
Therefore, when the FOTRIC 226B or 223B thermal camera is installed, we recommend that the temperature measurement route to be arranged in an "L" shape. After reaching the determined temperature measurement point, the channel turns left or right by 90°.
9. Face detection
This function needs to be activated online for first-time use. Then the network connection is no longer needed after activation. Please note: The network can be disconnected after activation, but do not disable the network, otherwise the software will prompt that it cannot connect to the visible light camera and the face detection function needs to be activated again. Please also make sure that the computer time is consistent with the local time.
10. Cannot display visible light image
There may be several reasons when the software prompts that the visible light (DC) image connection fails: 1. The USB cable connection of the visible light camera is faulty. Please reconnect or try again with a different USB port on the computer; 2. The computer may not have a camera driver installed. Please go online (Logitech official website: https://www.logitech.com), download and install the relevant driver. 3. Please check if the network is disabled. Disabling the network may cause the visible light image to fail.
11. Cannot display infrared thermal image
First, please make sure that the device 226Bor 223B is power on, and the status light is green. Second, make sure that the device driver is installed correctly. Third, make sure the USB connection between the device and the computer is stable.
12. Windows 8 / 10 version of the driver cannot be successfully installed
For the software WLIR-IR v220.127.116.1178 or newer version, the driver is digital signed and the step of "driver signature enforcement" is no longer needed! Please check whether you downloaded the right version.
Please check whether the driver signature enforcement is disabled (Disable driver signature enforcement) . If this step is not performed ,the prompt below will appear. The second possibility is that a 32-bit driver (x86) or a 64-bit driver (x64) is installed on a 64-bit or 32-bit system, respectively.
13. Live detection
Live detection is a new feature that can verify whether the measured object is a real living body, which can exclude the effects of photos, videos, and screen recordings, etc.
Please note that this function will be continuously upgraded and may have a certain error rate.
To manually shut down, press and hold the power button for about 6 seconds.
If the device is disconnected from the software (software has been closed or the computer is in sleep), the device will automatically shut down after 5 minutes.
15. Unable to detect faces
The face detection rate of the software cannot always be completely accurate. It is necessary to pay attention to these situations in which faces may not be successfully detected: people with black masks; people with long hair covering their foreheads; people with hats; people with less hair; people walking too fast, etc. Additionally, mis-identification may occur where other objects are mistakenly considered as human faces for temperature measurement.
16. Can the images not be saved?
You can go to system setting -> alarms -> uncheck the box for "Alarm snapshot" or "Face recognition snapshot".
17. Can it hide the digital image?
You can go to system setting and selece privacy mode. By this way, it will only display the infrared image without any snapshot.
18. Why do I get an error message "Invalid matching. ir image are not fully included in dc image." when I tried to match the IR and DC image?
This error message pops up when portion of IR camera field of view stands out of the scope of the digital image. The DC image position might be altered.
Stand in front the camera, check whether the DC camera lens is aligned well vertically with the IR camera lens, if not, move the DC camera toward left or right to make the close vertical alignment.
Get someone standing 2, 2,5, or 3m distance from the camera, remember this distance selection. By default it is 2m.
Rotate the DC camera up/down to so every part of the person shown on the IR image should also be seen on the DC image
There is a small arrow next to the menu "Match". Select the right distance and click "Match" to start the matching process.
Move the red dot and green dot on both IR image and DC image around the human head. Make sure the feature position on the IR image can match the the same feature position on the DC image. Such as the ear, top left face corner, neck, etc.
19. What's the minimum computer requirement?
Computer with windows 7 or 10 OS and
Core i5 7020 or above processor.
8GB or larger memory.
2x USB ports, USB 3.0 is better
More than 20GB of available hard disk space.
1024 x 768 display minimum, 1400 or higher is recommended.
Independent video card is recommended.
Keyboard and Microsoft mouse, or compatible pointer device.
Microsoft windows 7, 32/64 bit (you need to install. Net framework 4.0 or the program will not work normally), or
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.
Returns Our policy lasts 30 days. If 30 days have gone by since your purchase, unfortunately we can’t offer you a refund or exchange.
To be eligible for a return, your item must be in like-new condition and also be in the original packaging.
Several types of goods are exempt from being returned. Perishable goods such as food, flowers, newspapers or magazines cannot be returned. We also do not accept products that are intimate or sanitary goods, hazardous materials, or flammable liquids or gases.
Additional non-returnable items: - Gift cards - Downloadable software products - Some health and personal care items
To complete your return, we require a receipt or proof of purchase. Please do not send your purchase back to the manufacturer.
There are certain situations where only partial refunds are granted (if applicable) - Book with obvious signs of use - CD, DVD, VHS tape, software, video game, cassette tape, or vinyl record that has been opened - Any item not in its original condition, is damaged or missing parts for reasons not due to our error - Any item that is returned more than 30 days after delivery
Refunds (if applicable) Once your return is received and inspected, we will send you an email to notify you that we have received your returned item. We will also notify you of the approval or rejection of your refund. If you are approved, then your refund will be processed, and a credit will automatically be applied to your credit card or original method of payment, within a certain amount of days.
Late or missing refunds (if applicable) If you haven’t received a refund yet, first check your bank account again. Then contact your credit card company, it may take some time before your refund is officially posted. Next contact your bank. There is often some processing time before a refund is posted. If you’ve done all of this and you still have not received your refund yet, please contact us at order@NovaTestEquip.com.
Sale items (if applicable) Only regular priced items may be refunded, unfortunately sale items cannot be refunded.
Exchanges (if applicable) We only replace items if they are defective or damaged. If you need to exchange it for the same item, send us an email at order@NovaTestEquip.com to request exchanges for the defective item.
Gifts If the item was marked as a gift when purchased and shipped directly to you, you’ll receive a gift credit for the value of your return. Once the returned item is received, a gift certificate will be mailed to you.
If the item wasn’t marked as a gift when purchased, or the gift giver had the order shipped to themselves to give to you later, we will send a refund to the gift giver and he will find out about your return.
Shipping To return your product, send us an email at order@NovaTestEquip.com to request exchanges for the defective item.
You will be responsible for paying for your own shipping costs for returning your item. Shipping costs are non-refundable. If you receive a refund, the cost of return shipping will be deducted from your refund.
Depending on where you live, the time it may take for your exchanged product to reach you, may vary.
If you are shipping an item over $75, you should consider using a trackable shipping service or purchasing shipping insurance. We don’t guarantee that we will receive your returned item.